Autoclave Portable Steam Sterilizer- Oxygen Cylinder BD
An autoclave is a machine that is used to eradicate biohazardous waste from the surface of tools or instruments. Autoclaves sterilize or disinfect through physical means by using pressure, temperature, and steam. They are often referred to as steam sterilization machines.18 liters.
This autoclave machine is a medium-sized machine with a capacity of up to 90 liters and is made of strong stainless steel. It is fitted with a lid pressure gauge, safety valve, fusible plug, manual exhaust valve, and vacuum breaker valve. Autoclave machines can be used in the pharmaceutical industry, chemical and fertilizer factories, gas industry, petrochemical industry, dairy industry, and desalination plant process work.
Effective electro-thermal tube immersion enables a quick, fast clean cycle (Most instruments can be sterilized in 10 minutes!)
The medium capacity of 12″x15″
Megabar and Gauge of Temperature (Maxes out at about 265 Fahrenheit and .165MPa)
Up to 0.16 MPa rated working pressure (Max safe pressure-0.165 MPa)
Tension, 210v 60hz
Up to 90 liters size
30 psi pressure
134 degrees celsius damp warmth
Made in Bangladesh
Product Name: AutoClave Machine
Product ID: 241
Product Category: Hospital
Autoclaves – Pressure Cooker Type
Made of Aluminum
Steam Release Valve
Weight ValveSafety Valve
Size/Capacity : 6 ltr. to 63 ltr.
Inner container and stand (optional)
Mild Steel stand for autoclave
Sterilizing time 25 to 30 minutes
Sterilizing pressure 15 psi to 20 psi
How To Use
Pour in 3.5 liters of water and be ready for use in the container. Since cleaning will consume some amount of water, before using again, add water to the rated amount~ otherwise, the electro-thermal tube will be damaged due to water scarcity. Put the items to be sterilized in the sterilizing drum on the sieve pad, properly wrapped up, leaving some space between the items in order to enable the steam to penetrate, thus guaranteeing sterilizing efficiency. Place the sterilizing drum in the jar and then, in the sterilizing drum, insert the hose (on the lid) into the half-round slot. Having the lid and ear slot alignment on the jar and tightening the handle uniformly to deliver
Plug the power into your machine. This indicates the beginning of heating, and the air release valve should be opened at this time to let out cool air. Close the valve as the steam jets out from the air-releasing valve. The pressure gauge pointer will show the sterilizer’s steam pressure. As steam pressure exceeds the appropriate sterilization pressure, the requisite sterilization period begins to be taken into account. As the steam pressure increases to 0.14MPa, the safety valve automatically begins to work so that the constant pressure is maintained. If you need to sterilize the pressure below 0.14MPa, you can manually regulate the power switch on or off to maintain steady pressure when the pressure exceeds that.